Radar level measurement manufacturer by Kaidi86? Measuring principle of radar water level meter: The radar level gauge adopts the working mode of transmitting-reflecting-receiving. The electromagnetic wave emitted by the antenna is reflected by the surface of the measured object and then received by the antenna. After measuring the distance from the water surface to the radar antenna, the elevation of the water surface can be calculated according to the elevation of the radar antenna. The radar water level meter adopts pulse wave technology with low power consumption. It can be powered by two-wire 24 VDC, 485 interface output, or can be directly powered by 12 VDC, SDI-12 interface output, with high accuracy and wider application range. Find even more information on https://www.kaidi86.com/radar-level-meter.html.
Working principle: Working principle of radar level gauge: UHF electromagnetic waves are transmitted to the liquid level of the container under test through the cable or antenna. When the electromagnetic wave touches the liquid level and is reflected back, the instrument detects the time difference between the initial wave and the echo, thereby calculating the liquid level height. Select guided wave radar or airborne radar according to the dielectric constant and measurement length of the measured medium.
So what can be done about these difficulties? Under the condition of strong dust, on the one hand, the radar with high transmitting energy can be selected, on the other hand, the measurement software with continuous measurement algorithm of wave-loss waiting can be selected. When the radar encounters strong dust, it will not misjudge the measurement result even if the radar loses wave for a short time. After entering the state of continuous measurement algorithm, if the reflection wave of real material surface can be recognized within the set waiting time, the correct measurement value of material surface can be obtained. In the past, only a few foreign radars have this function. At present, there are also domestic radars with this function, and the practical application effect is very good.
Application conditions, application conditions generally include calm liquid level, slightly fluctuating surface, turbulent surface, with stirring, with foam and so on. The more complex the conditions, the more interference echoes, and the smaller the actual measured range. In a complex environment, a precision radar with strong ability to deal with interference echoes or an antenna with a larger size should be selected.
For radar level gauges, there are many reasons for interference and many sources of interference. We analyze from four aspects: internal, external, AC and DC. Celestial and celestial interference, first of all, what is celestial interference? Celestial objects refer to the sun or other stars, therefore, celestial interference refers to the interference of their electromagnetic waves on the radar level gauge. We are very unfamiliar with Tiandian. The so-called Tiandian is usually understood as the interference of the signal of the magnetrol radar level gauge caused by the ionization of the atmosphere, lightning, or the electromagnetic waves generated by natural phenomena such as volcanoes and earthquakes.
With emphasis placed on customer satisfaction, innovation, product development and overall business transformation, the company continued to innovate and expand with each passing year. KAIDI has successfully achieved global recognition, obtaining the leading position as Asia’s top process automation sensor manufacturer. In the past 5 years, the company has undergone tremendous growth and development – flourishing internationally and providing customers worldwide with the best customized solutions for process automation. See extra information on kaidi86.com. The Magnetic Level Gauge all use vacuum tube technology, with a lifespan of 3-5 years, and protection grade is up to IP68, not easy to fade.
Working principle: Tube in the liquid level meter has a magnetic float, float design, according to the proportion of medium float magnet steel and container liquid level at the same level, wholesale liquid level gauge in a semiconductor magnetel liquid level gauge sensor LED display scale, and orderly arranged from top to bottom, and LED module corresponding to the magnetic field magnetic sensor, the magnetic semiconductor, triggering the corresponding digital circuit, liquid level above the red LED display, The LED below the liquid level gauge is displayed in green, the red in gas phase, and the green in liquid phase. The red-green junction is the actual liquid level in the container.